For additional information, see the Global Shipping Program terms and conditions – opens in a new window or tab. To prepare for the loopback test, we wire together AI0 and AO0. This task was defined by me. At first I thought at using a shunt resistor in serie with my load, but then I show your solution. Then, you should test the individual channels of the USB to check that the input signals are detected correctly by the USB, and that the output signals generated by the USB have correct values. You can use the Index Array function on the Array palette.
|Date Added:||23 October 2011|
|File Size:||57.6 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Selecting physical channel for daw task. Sign in raq check out Check out as guest. If you know what an complex impedance is, you should be able to do the complex ni daq usb 6008 to match your setup. We will perform a very simple test, which is sufficient if we are to check that both AO0 and AI0 work correctly. Any international shipping and import charges are paid in part to Pitney Bowes Inc. This shows the physical channel associated with the virtual channel, see Figure Assume as an example ni daq usb 6008 your application requires two analog inputs.
measurement of resistance by USB – Discussion Forums – National Instruments
This item will be shipped through the Global Shipping Program and includes international tracking. Most Active Hardware Boards: Ni daq usb 6008 USB Table of contents: Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. Please enter 5 or 9 numbers daqq the ZIP Code. You must define one task for each of the input or output operations, i.
Report item – opens in a new window or tab. The Build Array function is used to construct this array, which in this example usn only one element.
This item will ship to United Statesbut the seller has not specified shipping options. In the window shown in Figure 16, we select ai0which opens the window shown in Figure Back to home page.
I use DAQ assistant and have to input data for Iex value. The video replaces the present document.
A quick guide to National Instruments USB-6008
However, the USB has several input channels and several output channels. I am trying to measure resistance by USB ni daq usb 6008 Contact the seller – opens in a new window or tab and request a shipping method to your location. Please enter a number less than or equal to 5. Output low is below 0. The present document should be sufficient to get started.
Interactive Control Panel for USB-6501, USB-6008, and USB-6009 using DAQmx Base
How can I determined the Iex value? Seller assumes all responsibility caq this listing. Delivery times may vary, especially during peak periods. The configuration can be saved using the Save button, see Figure A more ‘precise’ ni daq usb 6008 is possible if you apply some known resistors to determine more presice values of the build in resistors.
National Instruments USB Data Acquisition Card, NI DAQ, Multifunction | eBay
Other offers may also be available. Sell now – Have one to sell?
The VI was used to control this DC motor using analog output and to measure the tachometer voltage using analog input. Read more about the condition.